|Input Voltage||100/110/120/220/230/240 VAC,50/60Hz|
|Monopolar Output Power||130 W (MAX 100 -/+ 20%)|
|Power Consumption||450VA + 10% Under on maximum load|
|Bipolar Output Power||90 W (Max 50 -/+ 20%)|
|Brand||Dr Oppel Dual RF 4 MHz|
Dr. Oppels electorde is not being heated by itself. Electrode lets the cellular tissure produce the joule heat, and operates hemostasis, cutting and coagulation 4MHz radio frequency makes possible precise cutting and deep coagulation even in water. MOreover, coagulation depth is determined by electrode as 5mm,3mm, and 2mm.
Therer is no carbonization and damages to the surround cellular tissure by electric spark
and safety can be secured because this system is free from electric shock and bum which current electric surgical units make.
Make a hole in ear drum by transmilting the radio frequency on the top of the insulated cutting needle and remove inflammatory discharges. The hole sightly coagulated will be cured soon naturally.
Electrode Monopolar Bipolar Operation Time Frequency Timer Setting Output controlE01, E02Monopolar3-4minContinuous10F01-F05Monopolar10-20secContinuous10G,H seriesMonopolar10-20secContinuous10Bipolar ForcepsBipolar1-2secContinuous8-10Needle,LoopMonopolar-Continuous106mm BallMonopolar-Continuous103mm BallMonopolar-Continuous7-82mm BallMonopolar-Continuous6-7Spear,BladeMonopolar-Continuous10